HAMEG 203 6 PDF
Find great deals for HAMEG HM 20mhz Oscilloscope. Shop with confidence on eBay!. Oszilloskop HM Equipment HAMEG GmbH, Frankfurt, build , 12 pictures, 11 schematics, 1 tubes, Germany, semiconductors, Service- or Lab. Finally got the dust off my Hameg I bought from a car boot sale about 2 years ago and forgot about it. I didn’t have the space, then moved to a.
|Published (Last):||17 July 2017|
|PDF File Size:||10.56 Mb|
|ePub File Size:||4.86 Mb|
|Price:||Free* [*Free Regsitration Required]|
HAMEG Model Hm203-6 2 Channel 20 MHz Oscilloscope
The ones I’ve found are in German, and contain schematics only, no instructions. Someone gift me this oscilloscope and I fix some capacitators on it but I see this a bit weired.
I’m replacing the power line wired with a new one but will be in months untils I will have ground wires, I life in a old house with wires cover by sewing thread something like that. Should be fine to power it on and adjust it with the cover off, just be careful not to short anything with a metal screwdriver and stay away from the HV supply to the CRT.
Category:HAMEG – Wikimedia Commons
Z80user on December 04, Until now I fix some switch, create a new stop for one switch and replace 3 capacitors but now I don’t know what I can do with this noise. There was an error while thanking.
So my questions are, Is there anything else to it, besides the pot adjustment? I play with my fingers on multimeters too and saw the resistance between both hands is lower then between 2 fingers on hmaeg same hand if I active my muscles also I do it with radio and my body act like a antenna if I disconect it is like a flat line, I think the form of the wave form will be more like a sinwave.
Turn all three red variable controls fully counter-clockwise.
INP sockets are connected together and they are also connected to the mains earth when you get that in your house. I’m aware that exposed CRTs are lethally dangerous. The oscilloscope still a bit dush outside and on switchs, I don’t clear all of the switches.
The users who voted to close gave this specific reason: User manual here addition to previous link. Home Questions Tags Users Unanswered.
Pay also attention to the red variable controls. This question appears to be off-topic. Hameg weird 50Hz signal Does it look like a square wave 2 divisions high and 1 wide per period? Spend some time and clean the front panel and knobs and start playing with it. How to calibrate the vertical sensitivity of an old analog scope Hameg HM? The configuration now is two crocodrile wires with a resistor of 2K2 between both hhameg wires, this noise come mostly from the channel 1 on channel 2 I get some noise but a bit lower.
Oszilloskop HM Equipment HAMEG GmbH, Frankfurt, build
The problem is that hakeg no mention on how to do this in the manual I was able to find. Is it a terribly bad idea to do the adjustment while the scope is naked and powered?
If I mix both channels and add the inverse of the second one I can cancel this noise. I think the missing knobs can be replaced by something from the ELMA range. You can find the service manual here: Please login or register. As suggested, I just tried hzmeg and calibrated it myself. See it as a challange!
Z80user on December 03, Nothing scary, the scarier part was the amount of spider webbing inside: So I assume the “calibrated position” must be recalibrated. Jones play with the chair and see the spikes on the oscilloscope the noise of 50 Hz here is common but I think the form of the signal can be a bit 2033.